Working principle of Tubular Separator
Consist of the machine cover, driving device, drum, liquid catch tray, the liquid inlet and bearing seat, this separator with high running speed, belongs to sedimentation separator. The upper drum is a flexible shaft, while the lower is damping floating bearing. The main shaft is connected by a coupling buffer and a passive wheel. Motor transmit power from the transmission belt and tension wheel to passive wheel, to drive the drum rotate around its own axis with high-speed, and thus form a strong centrifugal force field. All the materials are fed from liquid inlet under the bottom, and flow upward along the drum wall by centrifugal force fluid. During this process, the dense liquid phase forms the outer ring, and the small liquid phase forms the inner ring,
which separately flow to respective liquid outlet mounted on the upper drum.
When the thickness of the sediment affect the liquid phase clarification or the capacity of the rated slag weight in the drum, the dregs of the drum wall should be removed manually after the shutdown.
Application：Tubular Separator is mainly used for separating all kinds of suspension which is difficult to be separated, especially suitable for separation of solid-liquid suspension with light concentration, heavy viscosity, fine particles and very small differential gravity of the two phases. such as clarification of herbs plants extracts, various medicine liquor, liquid medicine, oral liquid, the extraction of paint, dye, as well as various proteins, alga, fruit gels, and purification of rubber solution, coal tar, graphite, saponin, pectin, sirup, vaccine mycelia and all kinds of glucose; refining resin, and latex solution, enzyme (SOD) of oxide, various kinds of albumens, the glucoside of the soap, etc.
The main feature is the high rate of return of plasma and blood cells in the separation process of the crushing rate to a minimum.
Machine and material contact parts made of stainless steel, meet the GMP requirements of the pharmaceutical industry.Separation is widely used in humans, pigs, cattle, sheep, chickens and other kinds of blood plasma and blood cells
This machine has the advantages like: good separation effect,high yield,small cover area,easy operation etc.
Tubular centrifuge is mainly divided into clarify series GQ and separation series GF:
GF- separation type: mainly used for the separation of various difficult to separate the emulsion, especially for two-phase
density difference little liquid, liquid separation (difference in specific gravity greater than 0.1%), as well as a small amount
of liquid containing impurities, liquid, solid three-phase separator. Separation of various syrups;; separation of oil and the
like, transformer oil, edible oil dehydration and dewaxing, turbine oil, biodiesel, coal tar, lubricants, marine fuel oil,
quenching oil separation of whole blood - from whole blood the separation of plasma, blood cells; the separation of biological products; oil-water separation, sewage treatment.
Clarification GQ- centrifuge tube, mainly used for a variety of difficult to separate separating suspensions, especially suitable for low density, viscosity, fine solid particles, the solid-liquid specific gravity difference smaller solid-liquid separation.
Such as: large sugar oral liquid medicine, the floc more, fine solid particles and solid-liquid specific gravity difference is
small, and so on.